Accuracy of uploadable pedometers in laboratory, overground, and free-living conditions in young and older adults
© Dondzila et al.; licensee BioMed Central Ltd. 2012
Received: 8 November 2011
Accepted: 7 November 2012
Published: 11 December 2012
The purpose of this study was to examine the accuracy of uploadable pedometers to accurately count steps during treadmill (TM) and overground (OG) walking, and during a 24 hour monitoring period (24 hr) under free living conditions in young and older adults.
One hundred and two participants (n=53 aged 20–49 yrs; n=49 aged 50–80 yrs) completed a TM protocol (53.6, 67.0, 80.4, 93.8, and 107.2 m/min, five minutes for each speed) and an OG walking protocol (self-determined “< normal”, “normal”, and “> normal” walking speeds) while wearing two waist-mounted uploadable pedometers (Omron HJ-720ITC [OM] and Kenz Lifecorder EX [LC]). Actual steps were manually tallied by a researcher. During the 24 hr period, participants wore a New Lifestyles-1000 (NL) pedometer (standard of care) attached to a belt at waist level over the midline of the left thigh, in addition to the LC on the belt over the midline of the right thigh. The following day, the same procedure was conducted, replacing the LC with the OM. One-sample t-tests were performed to compare measured and manually tallied steps during the TM and OG protocols, and between steps quantified by the NL with that of the OM and LC during the 24 hr period. Mean error step scores (MES, criterion – device) and 95% Limits of Agreement (LoA) were calculated.
There were no significant differences between the OM and tallied steps for any of the TM speeds for either the young or older adult groups. The LC significantly underestimated steps for the young adult group during the 53.6 m/min TM speed (MES 31.4 [14.5, 48.3]) and during the OG < normal walking speed (MES 12.0 [0.9, 23.1] (p<0.01 for both age groups). The LC also significantly underestimated steps for the older adult group during the TM speeds of 53.6 m/min (MES 64.5 [45.6, 83.4]), 67.0 m/min (MES 15.1 [6.1, 24.0]), and 80.4 m/min (MES 3.2 [0.6, 5.9]) (p<0.01 for all speeds), in addition to the OG < normal walking speed (MES 14.7 [−13.3, 42.6] (p<0.01). The OM reported significantly lower steps during the 24 hr period for the young adult group by 949.1 steps (t=6.111, p<0.025) and for the older adult group by 612.9 steps (t=2.397, p<0.025).
Both the OM and LC pedometers were more accurate as TM and OG walking speed increased. The OM significantly underestimated steps during the 24 hr compared with a standard of care evaluation. Overall, both uploadable pedometers appear acceptable to use in young or old age groups to measure walking behavior.
KeywordsWalking Step-counter Precision Accuracy Motion sensor
Regular physical activity has long been shown to be host to a variety of benefits related to chronic conditions and diseases, such as diabetes, obesity, hypertension, and heart disease across a variety of populations . One commonly employed method in physical activity promotion is the utilization of pedometers, an inexpensive device that objectively monitors ambulatory physical activity. Numerous studies have documented beneficial health outcomes through pedometer-based physical activity interventions [2–4].
There are various styles of pedometers with differing technology to quantify steps, such as spring-levered and piezoelectric sensors. As the technology of these devices has advanced, newer pedometer models are able to connect to a computerized interface, such as a desktop computer, allowing ambulatory physical activity behavior to be uploaded. For instance, the Omron and New Lifestyle pedometer brands have such capabilities. Uploadable pedometers aim to further expand on the potential to increase and maintain activity habits to users by offering additional information and features. Such features could include individualized feedback and progress updates on daily walking behaviors, setting visual walking targets and how users compare to such targets, offering behavioral feedback cues based upon uploaded walking behaviors, all of which can be insightful during interventional purposes .
Numerous pedometer brands have been tested for their ability to quantify walking behavior, with the majority of such research focusing on younger, healthy adults [6–8]. Research examining the accuracy of such devices in the older adult population is more sparse [9–11], and collectively a paucity of data across all population ages exists evaluating newer technology uploadable pedometers. Therefore, the primary purpose of this study was to test the accuracy of two uploadable pedometers in measuring walking behavior during laboratory, overground, and free living activity in a group of community dwelling young and older adults.
A convenience sample of 102 adults participated in this study across two age groups: 20 – 49 years (n=53) and 50 – 80 years (n=49). All participants were recruited through word of mouth, posted fliers, and media announcements. Inclusion criteria consisted of being in general good health and able to participate in regular physical activity, and aged 20 – 80 years. Exclusionary criteria consisted of the inability to safely walk and/or run on a treadmill or around an indoor track, and/or the use of a required walking aid. One hundred and eleven individuals enrolled in the study, with 9 individuals dropping out. Of the 9 that dropped out (3 in the 20–49 years group and 6 in the 50–80 years group) the reason was an overall lack of time to finish all study visits. There were no demographic differences between those that finished the study and those that dropped out (data not shown). All participants were informed of potential risks and benefits of participation and signed an informed consent document approved by the University Institutional Review Board prior to study enrollment.
Participation in this study consisted of four separate visits. Visit one consisted of study explanation, obtaining informed consent, the completion of general demographics, and a treadmill walking protocol while wearing assigned pedometers for evaluation. During visit two participants were asked to complete an over-ground (track) variable speed walking protocol while wearing assigned pedometers. All participants were then asked to return for a third visit, and during this visit participants were instructed on how to wear the monitors for a 24 hour (24 hr) monitoring period. Finally, participants came back to return all pedometers following the 24 hr monitoring period, for a fourth and final visit. Each visit was separated by a minimum of 24 hours. All participants completed study visits in order.
Participants had their body height (to the nearest 0.1 cm) and body mass (to the nearest 0.01 kg) measured with no shoes and minimal clothing via a calibrated physician’s scale and stadiometer (Detecto, Kansas City, MO). Body mass index (BMI) was calculated by dividing body mass (kg) by height squared (m2). Waist circumference was measured in duplicate and averaged to the nearest 0.1 cm at the narrowest area of the trunk between the iliac crest and inferior rib using a tension-sensitive tape measure. Individual stride length was determined by a standardized process. Participants began with feet together and walked to a marker (103.6 m away). The distance was measured from the marker to the heel of the first foot that crossed the marker and added to the 103.6 m walked. Total distance walked was divided by the total number of steps taken to determine stride length. The accuaracy of steps quantified by the uploadable pedometers was examined across treadmill walking, over ground walking, and free living conditions.
Omron HJ-720ITC pedometer
The Omron HJ-720ITC pedometer (OM; Omron Corporation, Kyoto, Japan) is a waist-worn pedometer that uses a piezoelectric sensor to quantify steps and distance, including the ability to extrapolate caloric expenditure. The device is capable of storing up to 7 days of data for immediate retrieval in its display, and up to 41 days of data in its memory, which can be obtained by uploading the information to a computer. Information is automatically stored and reset at midnight each day.
Kenz lifecorder EX pedometer
The Kenz Lifecorder pedometer (LC; Suzuken Co. Ltd., Nagoya, Japan) is a waist-worn pedometer that uses a piezoelectric sensor to quantify steps and calories, as well as demarcate recorded steps between light, moderate, and vigorous intensities. The device is able store 7 days of information for immediate retrieval in its display, and up to 200 days of information stored in internal memory, which can be accessed by uploading the pedometer to a computer.
Treadmill walking protocol
Participants walked on a treadmill (TrackMaster TMX22, Newton, KS) at fixed speeds of 53.6, 67.0, 80.4, 93.8, and 107.2 m/min for 5-minutes at each protocol speed, or up until a point that participants reached 85% of estimated maximal heart rate. Each treadmill speed was verified with a digital tachometer (Shimpo Instruments, Itasca, IL) and found to be within ± 0.1%.
Prior to engaging in the treadmill walking protocol, each participant was fitted with the Kenz Lifecorder EX pedometer on the midline of the right thigh and the Omron HJ-720ITC pedometer on the midline of the left thigh, both secured to a belt at the level of the waist. Both anatomical site locations are supported and suggested by the manufacturers. During the treadmill walking protocol actual steps were tallied by a researcher using a hand-tally counter. In between walking protocol speeds participants straddled the treadmill, so that pedometer steps could be recorded from both the LC and the OM. This permitted a pre and post walking step count to be recorded from the LC and OM, to yield steps accumulated during each protocol speed.
Overground walking protocol
On a separate day participants returned to complete a track variable speed walking protocol. Each participant walked once around an indoor track (394 m) at three different self-determined speeds. These speeds were required to be < normal, normal, and > normal walking speeds. Hence, these speeds were variable across participants and self-selected. During this time each participant wore the LC and OM pedometers in midline of the right and left thigh, respectively, affixed to a belt at the level of the waist. During each track walking speed, total distance was recorded to calculate speed, and actual steps were manually tallied by a researcher. For all participants tested, walking speed increased across the three variable conditions of < normal, normal, and > normal. These speed allocations were designed to have each individual walk across their own self-determined walking speed range that may be typical of overground walking for them in a naturalistic lifestyle setting.
24 Hour monitoring period
A random subset of 20 participants agreed to engage in a 24 hr evaluation of each uploadable pedometer. Participants were instructed to wear pedometers for 24 hours, except when sleeping, and when in contact with water (such as bathing, showering, or swimming). For one 24 hr observation period participants wore the LC on the right midline of the thigh (same position as the laboratory testing) and on the left midline of the thigh participants placed a New Lifestyles NL-1000 (NL) pedometer (New Lifestyles, Inc., Warminster, PA), both secured on a belt at waist level. Although no gold-standard device exists to measure accrued steps on a daily basis under free-living conditions, the NL series pedometer is an industry standard pedometer, heavily utilized for interventional purposes. The NL series pedometer has an extensive empirical backing in the literature for validity and reliability, able to accurately quantify increasing walking intensity activities (<2% error) , while retaining high intramodel reliability (0.99) . Accordingly, NL pedometers have been shown to be substantially more sensitive to walking behavior during observational periods, compared to other pedometers . As such, the NL was worn as a standard of care comparison, not a criterion comparison, with the LC. Each participant was given instructions to write down day starting steps on the pedometer, and day ending steps on each pedometer. The difference constituted daily steps from each brand, the LC and the NL. The following day, this was repeated, with the OM pedometer worn on the left midline of the thigh, (same position as the laboratory testing) and on the right midline of the thigh participants wore the NL pedometer, both secured by a belt at waist level. Schneider et al.  and others  have shown high correlation coefficients (r=0.99) for NL brand pedometers when worn on the left and right side of the body, thus justifying the use of the NL-1000 pedometer to be worn on both the left and right side across days.
All statistical analyses were performed utilizing SPSS 19.0 for Windows (Chicago, IL). Descriptive statistics are expressed as mean ± standard deviation. For each treadmill walking and overground walking activity an error score was computed for each participant, by subtracting the estimate (pedometer) from the criterion (manually tallied) and compared with zero. Error scores of zero would indicate that there was no difference between the pedometer and criterion measure. Positive error scores represent underestimates, and negative error scores represent overestimates. The MES scores for each treadmill walking and overground walking speed for both uploadable pedometers were tested using one-sample t-tests and Bonferroni corrections for multiple comparisons. Limits of agreement (LoA) are presented as +/− 1.96 Standard Deviations from the MES. For the 24-hour observation the NL-1000 served as the standard of care comparison against the LC and OM. MES and 95% LoA were again calculated and statistically tested using one-sample t-tests.
Physical and descriptive characteristics of the participants (Mean±SD)
20-49 yrs (n=53)
50-80 yrs (n=49)
Stride length (cm·step-1)
Pedometer accuracy: treadmill walking
Mean error step scores and limits of agreement during treadmill walking protocol
OM: 20–49 yrs
−5.4 (−13.4, 2.5)
−2.7 (−6.4, 1.0)
0.8 (−2.9, 4.4)
−6.6 (−17.7, 4.5)
8.4 (0.2, 16.6)
OM: 50–80 yrs
9.7 (−4.9, 24.3)
−4.5 (−18.2, 9.1)
−0.2 (−2.2, 1.8)
1.9 (−0.9, 4.7)
12.4 (−1.3, 26.2)
LC: 20–49 yrs
31.4* (14.5, 48.3)
3.2 (−0.3, 6.7)
0.5 (−0.5, 1.6)
−1.2 (−5.4, 3.0)
0.3 (−0.6, 1.1)
LC: 50–80 yrs
64.5* (45.6, 83.4)
15.1* (6.1, 24.0)
3.2* (0.6, 5.9)
0.3 (−2.6, 3.1)
−0.2 (−1.5, 1.1)
Pedometer accuracy: overground walking
Mean error step scores and limits of agreement during overground walking protocol
<Normal walking speed
Normal walking speed
>Normal walking speed
OM: 20–49 yrs
OM: 50–80 yrs
LC: 20–49 yrs
LC: 50–80 yrs
Pedometer accuracy and precision: 24 hr observation
Mean error step scores and limits of agreement during 24 hour observation period
20-49 yrs (n=10)
50-80 yrs (n=10)
949.1* (597.8, 1300.4)
612.9* (34.4, 1191.4)
−305.1 (−709.9, 99.7)
−38.0 (−936.6, 860.6)
There is a plethora of research that demonstrates pedometers’ ability to promote increases in ambulatory activity. Efforts have been made to make pedometers more user friendly, such as reducing the user’s need to frequently record daily steps. Uploadable pedometers aim to achieve such capabilities, and also have the potential to provide the user with continual and individualized feedback. There is limited research on the accuracy of such pedometers, and this lack of knowledge resonates more so in the older adult population. The results of the current study showed that the OM and LC were increasingly accurate in their ability to quantify steps in both participant age groups as both treadmill walking speed and overground walking speed increased. The LC significantly underestimated steps during the 53.6 m/min stage for the younger adult group, and during the 53.6, 67.0, and 80.4 m/min stages for the older adult group. The LC also significantly underestimated steps for the younger adult group during the less than normal walking speed in the overground walking trial. During the 24 hr observation period, the OM pedometer significantly underestimated steps for both young and older adult groups.
Similar to previously published research, the OM pedometer generally became increasingly more accurate as walking speed increased on the treadmill. Using the same treadmill speeds as the current study, the Omron HJ-105 has been shown to have the largest percent error in measuring steps at 2.0 mph, with increasing accuracy as treadmill speed increased . A study by Foster et al. showed the Omron HF-100 to be accurate (>98%) in quantifying steps at speeds greater than 2.0 mph . Likewise, studies examining the validity of the Omron HJ-122, Omron HJ-720ITC, and Omron HJ-113 have demonstrated the pedometers’ high accuracy for quantifying steps at increasing speeds above 2.0 mph [18, 19]. These results, collectively with those of the current study, indicate that Omron pedometers become increasingly accurate in assessing ambulatory activity above slow walking speeds. One study, however, indicates otherwise. Crouter and colleagues showed the Omron HJ-105 to significantly overestimate steps at treadmill walking speeds at 4.0 mph (the fastest speed of the protocol) . The trend for increasing accuracy of the Kenz Lifecorder during increasing treadmill speeds resonates that of previous research, which shows the slowest walking speeds to have the largest mean error [6, 20]. Current study results also show significant underestimations at the 2.0 mph stage, but were much more evident in older adults. Although two additional walking speeds significantly underestimated steps in older adults in the current study, the corresponding percent errors were similar to previously reported results in younger adults . Overall, both the Omron HJ-720ITC and Kenz Lifecorder examined in the present study represent suitable options for walking behaviors assessed via treadmill, with minor decrements in accuracy at slower walking speeds.
The results of the current study for overground trials provides further credence that the pedometers examined in the current study are generally accurate in their measurements of walking activity. The OM of the current study reported more accurate step measurements than that of reported by Schneider et al., who reported an underestimation of 19.0 steps over a distance nearly identical to that of the current study . Even at the slowest walking cadence of the present protocol, the largest mean error was an overestimation of 4.7 steps (for the young adult group). These results are similar with those presented by Holbrook et al. . They assessed the validity of two Omron brand pedometers over 100 m walking trials, reporting absolute percent errors less than 2% for the HJ-151 for their slow, moderate, fast, and self-selected speed trials, and less than or equal to 2% for the HJ-720ITC for the same trials. The OM in the current study had percent errors less than 2% for the < normal walking speed, and less than 1% for the normal and > normal walking speeds. Currently, there exists limited research on the validity of the Kenz Lifecorder during overground walking. We are aware of one study that assessed such during self-selected walking paces on a track, which showed a less than 1% error in steps , which are very similar to the normal walking speed results of the current study. Overall, the OM and LC pedometers exhibited increasing accuracy during faster walking cadences during the OG protocol, a trend similar to that of the TM protocol.
The OM pedometer recorded less steps compared with the NL standard of care comparison in both the young age group and the older age group during the 24 hr observation period. Silcott et al. also examined the Omron HJ-720ITC pedometer during day long observation periods in a sample of adults aged similar (mean 31.3-46.2 years) to that of the young adult category in the present study (20–49 years), and showed the pedometer to significantly underestimate steps . Collectively, their results are likely due to the hardware of the Omron pedometer, as they only quantify steps after movement of four seconds or more. The magnitude of difference in error was approximately 13% for the young age group and 7% for the older age group in the current study. Although such differences are marginal, it highlights that a degree of caution is needed when comparing daily values of accrued steps across different pedometer brands.
There are several limitations of the current study that warrant mention. The populations used were all generally healthy, so it may not be appropriate to extend the results previously stated to diseased, those with gait impairments, or obese populations. Due to feasibility, the NL was used as the comparison variable for comparing OM and LC measured steps during the observation period rather than employing a manually tallied count. The current study does, however, fill an important void in the literature, by examining the validity of uploadable pedometers in both young and old age groups across laboratory, overground, and free-living activities.
Both the OM and LC pedometers were more accurate as TM and OG walking speed increased. The OM significantly underestimated steps during the 24hr monitoring period compared with a standard of care evaluation, and highlights that caution is needed when comparing total accrued step per day values across different pedometer brands. Overall, both uploadable pedometers appear acceptable to use in young or old age groups to measure walking behavior.
Omron HJ-720ITC pedometer
Kenz Lifecorder EX pedometer
This work was partially supported by a Career Development Award (Strath) from the National Institute of Aging (K01AG025962).
- Haskell WL, Lee IM, Pate RR, Powell KE, Blair SN, Franklin BA, Macera CA, Heath GW, Thompson PD, Bauman A: Physical Activity and Public Health: Updated Recommendations for Adults From the American College of Sports Medicine and American Heart Association. Circ. 2007, 116: 1081-1093.View ArticleGoogle Scholar
- Moreau KL, Degarmo R, Langley J, McMahon C, Howley ET, Basset DR, Thompson DL: Increasing daily walking lowers blood pressure in postmenopausal women. Med Sci Sports Exerc. 2001, 33: 1825-1831. 10.1097/00005768-200111000-00005.View ArticleGoogle Scholar
- Talbot LA, Gaines JM, Huynh TN, Metter EJ: A home-based pedometer-driven walking program to increase physical activity in older adults with osteoarthritis of the knee: a preliminary study. J Am Geriatr Soc. 2003, 51: 387-392. 10.1046/j.1532-5415.2003.51113.x.View ArticleGoogle Scholar
- Schneider PL, Bassett DR, Thompson DL, Pronk NP, Bielak KM: Effects of a 10,000 steps per day goal in overweight adults. Am J Health Promot. 2006, 21: 85-89.View ArticleGoogle Scholar
- Richardson CR, Buis LR, Janney AW, Goodrich DE, Sen A, Hess ML, Mehari KS, Fortlage LA, Resnick PJ, Zikmund-Fisher BJ, Strecher VJ, Piette JD: An Online Community Improves Adherence in an Internet-Mediated Walking Program. Part 1: Results of a Randomized Controlled Trial. J Med Internet Res. 2010, 12: e71-10.2196/jmir.1338.View ArticleGoogle Scholar
- Crouter SE, Schneider PL, Karabulut M, Bassett DR: Validity of 10 Electronic Pedometers for Measuring Steps, Distance, and Energy Cost. Med Sci Sports Exerc. 2003, 35: 1455-1460. 10.1249/01.MSS.0000078932.61440.A2.View ArticleGoogle Scholar
- Schneider PL, Crouter SE, Bassett DR: Pedometer Measures of Free-Living Physical Activity: Comparison of 13 Models. Med Sci Sports Exerc. 2004, 36: 331-335. 10.1249/01.MSS.0000113486.60548.E9.View ArticleGoogle Scholar
- Kang M, Bassett DR, Barreira TV, Tudor-Locke C, Ainsworth B, Reis JP, Strath S, Swartz A: How many days are enough? A study of 365 days of pedometer monitoring. Res Q Exerc Sport. 2009, 80: 445-453. 10.5641/027013609X13088500159129.View ArticleGoogle Scholar
- Cyarto EV, Myers AM, Tudor-Locke C: Pedometer Accuracy in Nursing Home and Community-Dwelling Older Adults. Med Sci Sports Exerc. 2004, 36: 205-209. 10.1249/01.MSS.0000113476.62469.98.View ArticleGoogle Scholar
- Grant PM, Dall PM, Mitchell SL, Granat MH: Activity-monitor accuracy in measuring step number and cadence in community-dwelling older adults. J Aging Phys Act. 2008, 16: 201-214.Google Scholar
- Storti KL, Pettee KK, Brach JS, Talkowski J, Richardson CR, Kriska AM: Gait Speed and Step-Count Monitor Accuracy in Community-Dwelling Older Adults. Med Sci Sports Exerc. 2008, 40: 59-64.View ArticleGoogle Scholar
- Schneider PL, Crouter SE, Lukajic O, Bassett DR: Accuracy and Reliability of Pedometers for Measuring Steps over a 400-m Walk. Med Sci Sports Exerc. 2003, 35: 1779-1784. 10.1249/01.MSS.0000089342.96098.C4.View ArticleGoogle Scholar
- Crouter SE, Schneider PL, Bassett DR: Spring-Levered versus Peizo-Electric Pedometer Accuracy in Overweight and Obese Adults. Med Sci Sports Exerc. 2005, 37: 1673-1679. 10.1249/01.mss.0000181677.36658.a8.View ArticleGoogle Scholar
- ACSM’s Guidelines for Exercise Testing and Prescription, Eighth edition: 2010, Philadelphia, PA: Lippincott Williams & Wilkins
- Samson MM, Crowe A, de Vreede PL, Dessens JAG, Duursma SA, Verhaar HJJ: Differences in gait parameters at a preferred walking speed in healthy subjects due to age, height and body weight. Aging Clin Exp Res. 2001, 13: 16-21.View ArticleGoogle Scholar
- Le Masurier GC, Lee SM, Tudor-Locke C: Motion Sensor Accuracy under Controlled and Free-Living Conditions. Med Sci Sports Exerc. 2004, 36: 905-910.View ArticleGoogle Scholar
- Foster RC, Lanningham-Foster LM, Manohar C, McCrady SK, Nysse LJ, Kaufman KR, Padgett DJ, Levine JA: Precision and accuracy of an ankle-worn accelerometer-based pedometer in step counting and energy expenditure. Prev Med. 2005, 41: 778-783. 10.1016/j.ypmed.2005.07.006.View ArticleGoogle Scholar
- Hasson RE, Jaller J, Pober DM, Staudenmayer J, Freedson PS: Validity of the Omron HJ-112 pedometer during treadmill walking. Med Sci Sports Exerc. 2009, 41: 805-809. 10.1249/MSS.0b013e31818d9fc2.View ArticleGoogle Scholar
- Giannakidou DM, Kambas A, Ageloussis N, Fatouros I, Christoforidis C, Venetsanou F, Douroudos I, Taxildaris K: The validity of two Omron pedometers during treadmill walking is speed dependent. Eur J Appl Physiol. 2011, epub ahead of printGoogle Scholar
- Abel MG, Hannon JC, Sell K, Lillie T, Conlin G, Anderson D: Validation of the Kenz Lifecorder EX and ActiGraph GT1M accelerometers for walking and running in adults. Appl Physiol Nutr Metab. 2008, 33: 1155-1164. 10.1139/h08-103.View ArticleGoogle Scholar
- Holbrook EA, Barreira TV, Kang M: Validity and Reliability of Omron Pedometers for Prescribed and Self-Paced Walking. Med Sci Sports Exerc. 2009, 41: 669-673.View ArticleGoogle Scholar
- Silcott NA, Bassett DR, Thompson DL, Fitzhugh EC, Steeves JA: Evaluation of the Omron HJ-720ITC Pedometer under Free-Living Conditions. Med Sci Sports Exerc. 2011, 43: 1791-1797. 10.1249/MSS.0b013e318212888c.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.